For the modern and complex electronics industry working with surface coatings becomes more and more a basic requirement. 3D MID (Molded Interconnect Device) is based on every conceivable free geometric shape in order to incorporate solutions for robotic fingers or portable electronics in textiles. Glass coatings are used for highly complex multi-layer display solutions to add even more functionality. However, all these components have one thing in common, they need a long service life and be resistant to various environmental influences.
In order to ensure this, destructive material testing is usually performed. The most well-known are scratch-, stud pull tests. Our September 2015 newsletter described the scratch test. In this newsletter we now describe a practical solution developed by XYZTEC to perform a very reproducible stud pull test. The stud pull test is based on the principle that an adhesive connection is made with the carrier (or object to be tested) by using a glue that is stronger than the bond that needs to be tested.
Two of the biggest challenges are to get an even distribution of the glue and the long curing time associated with some adhesives. If, due to uncontrolled application, the distribution of the glue is not even, menisci are formed that can have a significant influence on the standard deviation and make reproducibility almost impossible. This explains why the stud pull test is only used in limited cases and not during production. The positioning of the studs and the vertical force required to pull the studs also have a negative influence.
The solution developed by XYZTEC (in cooperation with other partners) has looked at all these problems and found a solution. The Condor Sigma tester was provided with a micro dispenser that can dispense very small drops of glue (micro liter amounts) very accurately and is cured with UV light. Combined with a USB Tweezer and special tips that look like the inverse of a mushroom, a glass stud can be taken from a magazine fully automatically and placed with a predefined pressure on the surface with the glue. The push force is just sufficient to get a strong connection, without forming any menisci. A special UV light that has been mounted on the machine cures the glue within seconds. The tweezer holds the stud until the test is performed. Before the test the tweezer is positioned in such a way that no unwanted load is applied on the stud, to avoid any influence on the measurement.
When the tweezer is released and the operator manually moves the stages, there is a very real risk that the operator due to carelessness bumps into the stud. This applies an unwanted load that can have a significant influence on the test result. With the above method this is nearly impossible. Because UV light cannot be used for all applications XYZTEC also offers different solutions with other studs and glue. Please contact us for more information if this article has aroused your interest, or if you have any questions.
|2019年7月||Meet the world’s biggest bond testers|
|2019年4月||Free SMTconnect ticket? Register now|
|2018年8月||Is Fully Automatic Bond Testing Possible?|
|2017年7月||Advances in Thin, 3D and MEMS Die Bond Strength Testing|
|2017年5月||Shear testing copper pillar|
|2017年3月||Fracture strength of thin wafers and die|
|2017年2月||Centralized database for enhanced security and SPC options|
|2017年1月||Full Automation with Improved Pattern Recognition, Fiducial Marks Analysis, Wire Detect and Failure Mode Analysis|
|2016年12月||Coating and film testing|
|2016年11月||Latest software release enables automatic grading|
|2016年10月||Vision enhanced materials testing|
|2016年9月||XYZTEC makes lid pull easy|
|2016年8月||Condor Sigma W12 with large heater stage|
|2016年7月||How to: Wire Pull|
|2016年6月||Get 10% off your next tools order|
|2016年5月||Pull testing bumped wafers|
|2016年4月||SMD gull wing leads testing|
|2016年3月||Automation: matrix bond testing made easy|
|2016年2月||Fully integrated solution for automated wafer testing|
|2016年1月||In bond testing, size does matter|
|2015年12月||XYZTEC goes from strength to strength|
|2015年11月||Automatic wire detect|
|2015年10月||Highly reproducible stud pull tests are possible|
|2015年9月||刮痕测试 / 镀膜测试|
|2015年8月||Loading samples has never been so easy|