Home » KnowledgeCenter » Pull tests » Tweezer pull
Tweezer pull is simply a pull test performed by a tweezer. During the test an upward force is applied to a wire, a ribbon or some other feature on your sample after being gripped by the tweezers. In many cases, the sample has to be prepared for testing first.
A peel test is essentially the same, but apart from an upward movement, the XY stage also does a controlled move. Normally, the purpose is to maintain a perpendicular force on the ribbon or wire being pulled off the substrate.
We aim to spread knowledge about bond testing in the industry. That is why we have an extensive “how-to” on Tweezer Pull on our website.
Tweezer pull testing is comparable to standard pull testing. The difference between them is that special tweezers are used to clamp the part and hold it before pulling the part. Tweezer pull testing is used to perform pull tests on wires, ribbons, pins, balls or components. It is important that the part that is tested is not dislocated when the tweezer is closed.
For a standard wire pull test the hook applies the force by pulling the wire. The applied force on the wire is not equal to the force applied to the ball or wedge, because the force is not applied exactly above this ball or wedge. With a tweezer pull test it is possible to apply the force directly above the ball or wedge. The measured force will equal the applied force. Before performing this test, it may be necessary to cut the wires first. This can be done with special tips that can cut the wire. It is also possible to shear the bond on one side, straighten the wire with the tweezer and then perform the test.
Xyztec offers the USB tweezer as a general solution for most tweezer pull tests. The tips on this tweezer can be changed and the required force to clamp a part can be set in the software. We also offer special tips to cut a wire, and we have specialized tweezers for higher forces or very small parts.
During use, solder builds up in the cavity in cold bump pull (CBP) jaws. This reduces their grip efficiency and affects the quality of your measurements. Cleaning the cavities is very difficult and small ball cavities can be damaged easily.
Click here to read the article that was first published in Chip Scale Review May June 2014.
Click here to download an article by Faulhaber, called “Mechanical stress test for component solder joints and bonding wires – Micro drives with the ‘strength of a giant'” (PDF) on the USB Tweezers.
Local support and parts, global presence, world-class testers
J.F. Kennedylaan 14b5981 XC PanningenThe Netherlands
Schäferei 1806237 LeunaOT GünthersdorfGermany
72/7 M.12 Soi. SoonthornwipakBangpla, Bangphli,10540 Samut PrakanThailand
No. 157, Zhongzheng 6th St., Hukou Township, Hsinchu County 303, Taiwan (R.O.C.)
Room 2012 Haichuang Mansion, No.288 Dengyun Road, High-tech district, Kunshan, Jiang Su, China
Click to download thispage as PDF.
Follow us on:
© 2021 xyztec bv
Xyztec develops world-class bond testing technologies and works together with global partners to provide local support worldwide.
Necessary cookies are absolutely essential for the website to function properly. These cookies ensure basic functionalities and security features of the website, anonymously.
Analytical cookies are used to understand how visitors interact with the website. These cookies help provide information on metrics the number of visitors, bounce rate, traffic source, etc.
Advertisement cookies are used to provide visitors with relevant ads and marketing campaigns. These cookies track visitors across websites and collect information to provide customized ads.